ENRAF 854 ATG MANUAL PDF
ENRAF-NONIUS ATG Interface. Tank Gauging Protocols Manual. ENRAF INTRODUCTION. The interface between BBI’s Network // Honeywell Enraf ATG Pdf User Manuals. View online or download Honeywell Enraf ATG Installation Manual. Honeywell Enraf series ATG servo gauge is a reliable, versatile and accurate automatic tank level MB; Honeywell Enraf Servo Gauge ATG Manual.
|Published (Last):||17 November 2012|
|PDF File Size:||15.11 Mb|
|ePub File Size:||19.87 Mb|
|Price:||Free* [*Free Regsitration Required]|
One revolution is divided into steps, therefore one step is equivalent to 0.
Enraf servo ATG install guide |
A Note points out a statement deserving more emphasis than the general text, but does not deserve a “Warning” or a “Caution”. S1 Check whether the set point 1 is set correct; ernaf not, change it. Use the following procedure for calibration. Unlock Give an unlock command to cancel the calibrate command.
Enter the position of the top of the ball valve with respect to the tank zero adding the immersion depth of the displacer at the product level. Device Cable requirements Spot temperature element: Place shark rings between flange and strip figure 4. The electronic compartment contains the following sub-assemblies refer to emraf 5.
Besides the error codes, data items from level and wnraf functions such as temperature and analog level output contain one or more status bytes which also give valuable information. Other features such as: The gauge will now initialize and after start-up, the modified settings become active.
Repeatability test If the test magnet item 7 of the tool kit is hold for approximately 5 seconds in front of the rectangle block below the display, a repeatability test is performed. Remove the two screws holding the force transducer by lifting it slightly in order to release the span wire from the Figure 5. Warning Do not use the instrument for anything else than its intended purpose.
Request for the maximum unbalanced weight. That can be calculated as follows refer to figure 3.
Enraf servo 854ATG install guide
This is a safety grounding requirement. For high pressure version max. During the test, the level dimension and level type on the display are replaced by exclamation marks!!! Enter the position of the top of the ball valve with respect to the tank zero adding the immersion depth of the displacer at the product level. Alternatively, the Honeywell Enraf service tool Ensite can be used to configure the instrument.
Step 3 Check with the identification code on the label of your ATG whether the gauge is equipped with one or more optional functions, and program the items for that options refer to the appropriate option manuals.
If a discrepancy exists between measured and desired value, an advanced software control module adjusts the position of the stepper motor. Honeywell Enraf series ATG servo gauge is a reliable, versatile and accurate automatic tank gauge with a minimum of moving parts, designed to measure all kinds of liquids in any type of storage tank.
Detected errors will be shown as status information on the display refer to Appendix Dor can be requested as items by the PET. DA Check whether the displacer area is set correct; if not, change it. The reply on the PET display will be: Page 44 Trouble shooting 6. The voltage selector is located inside electronic compartment on the top of backplane.
Check that non-used cable inlets are sealed with appropriate stopping plugs. This command must be repeated after each EX exit or RS reset command. Maintenance Page 42 5.
Instruction manual series ATG level gauge
Solve the problem or adjust item ML. Program frequency 2, measured with test weight of g. Issue a FR freeze command when the displacer comes within reach. Open the electronic compartment cover front cover. Refer to the below table: